Raman electrochemical flow cell setup
This cell combines two classical analytical techniques such as electrochemistry and Raman spectroscopy, to obtain in-situ chemical information about the reactions taking place during an electrochemical experiment. It consists mainly of three elements: (i) sample holder with copper foil serving as an electric contact to the working electrode surface, (ii) chamber in which reference and counter electrodes are installed together with electrolyte inlet and outlet and (iii) lid which seals the cell. The sample consisting of a thin film of electrochemically active material deposited on a rigid or flexible substrate (working electrode) is loaded from the bottom via magnetic (or screw) mount. The counter or/and reference electrodes are mounted in a top casing (either 2-, or 3-electrode setup. The oval counter electrode made of e.g. Platinum wire assures uniform distribution of the field lines along the path to the working electrode. During an experiment, Raman laser is focused onto the surface of a thin film working electrode through a transparent sapphire window and thin layer of electrolyte (total optical path of 3.25 mm). The electrolyte thickness of 2.25 mm ensures free diffusion of ions (e.g. protons) and its counter ions.
The cell elements are constructed with materials that are inert to the sample (glass and PEEK). It well fits aqueous (FKM/EPDM O-Rings) and organic solvent (FFKM O-Rings) electrolyte requirements. The construction is gas-tight and can be used when the removal and exclusion of contaminants such as oxygen and water is required by bubbling of an inert gas through the electrolyte (in an external reservoir).
This cell can be used to track kinetic phenomena such as the near-surface proton concentration changes during oxidation and reduction reaction at working electrode. It can be also used to identify materials such as carbon, metal oxides, polymers and electrolytes, and to determine their structure and distribution. Various metals are suitable for this cell as auxiliary electrode including Platinum, Gold and Silver.
Operating Video Manual:
nominal exposure area: 3.5 cm2
electrolyte volume: 4.5 mL
optical path (including sapphire window): 3.25 mm
electrode plug diameter: 6 mm
minimum sample size: 23 mm x 23 mm
recommended sample size: 25 mm x 25 mm
1 x Magnetic Mount Raman Electrochemical Flow Cell, Active Area: 3.5 cm2, volume: 4.5 mL – MM Raman ECFC 3.5 cm2, 4.5 mL (or Raman Electrochemical Flow Cell, Active Area: 3.5 cm2, volume: 4.5 mL – Raman ECFC 3.5 cm2, 4.5 mL)
1 x Metal Wire Auxiliary Electrode (Platinum)
1 x Reference electrode (Ag/AgCl, non-aqueous Ag/AgCl)
We Also Recommend