IKA VACSTAR Control Vacuum Pump
Buy IKA products from MSE Supplies at the best value.
MSE Supplies is an authorized distributor of IKA.
Item number: 0020109376
Separation, filtration, removal: The VACSTAR dry 4-chamber membrane vacuum pump is powerful, resistant to chemicals and low-maintenance. VACSTAR control is the ideal system with its integrated vacuum controller and removable display. The controller offers a suitable mode for the desired process: from automatic to manual operation, right through to pumping or an evacuation and cleaning function.
VACSTAR has an excellent suction performance, with its 22 l/min at a minimum of 2 mbar.
Integrated vacuum controller with removable controller (WiCo)
Depending on the application, you can easily select your operating mode either by using the intuitive operator menu or by remote control.
For additional safety for users and environment, we recommend using the VSE 1 vacuum safety emission condenser, which prevents solvents from being released into the room air. The VSE 1 is available as accessory.
Scope of delivery
- WiCo VACSTAR control
- Vacuum hose (1 m)
- OS 1.0 WiCo power supply unit
- USB Cable - USB A to Micro-B, 2 m
- USB A - USB B cable
- Screw driver
Founded in 1910 as a supplier for pharmacies and hospitals, the company left bombed-out Cologne for Staufen im Breisgau in 1942. There it quickly became the world's leading company for laboratory technology as well as dispersing, stirring and kneading machines.
Technical Data of IKA VACSTAR Control Vacuum Pump
|Pump rate max. (50/60Hz)||1.32 m³/h|
|Pump rate max. (50/60Hz)||22 l/min|
|Final pressure without gas ballast||2 mbar|
|Connection diameter suction side||8 mm|
|Connection diameter pressure side||8 mm|
|Input pressure||2 - 1030 mbar|
|Boiling point detection||yes|
|Analog speed vacuum control||yes|
|Speed adjustment||Turning knob|
|Speed range||285 - 1200 rpm|
|Pressure unit / scale||mbar, hPa, mmHg, Torr|
|Vacuum sensor type||ceramic Al2O3|
|Pressure max. for pressure sensor||1.2 bar|
|Measurement range (absolute)||1 - 1030 mbar|
|Control range||1 - 1030 mbar|
|Resolution pressure||1 mbar|
|Measurement uncertainty||1 mbar|
|Noise at low. pressure||54 dB(A)|
|Material in contact with medium||Al2O3, PTFE, FPM, PPS|
|Housing material||alu-cast coating / thermoplastic polymer|
|Mode pump %||yes|
|Dimensions (W x H x D)||270 x 435 x 500 mm|
|Permissible ambient temperature||5 - 40 °C|
|Permissible ambient conditions||80% (up to 31°C), decreasing linearly to max. 50 (@40°C)|
|Protection class according to DIN EN 60529||IP 20|
|RS 232 interface||yes|
|Voltage||100 - 240 V|
|Power input||140 W|
|Power input standby||1.5 W|